Author: Jason Herrington
Published By: Restek Corporation
Year of Publication: 2017
The NJ low level TO-15 air testing method is based on U.S. EPA Method TO-15, but it has more some differences that make it quite challenging. The broader calibration range (0.2–40 ppbv) is particularly difficult to meet. In this application note, we focus on the method’s analytical requirements and demonstrate how to meet them using a Markes Unity with CIA Advantage and a 30 m Rtx-VMS column.