Your web browser will no longer be supported by Restek.com as of 30 June 2021.
To avoid any interruption in access or functionality, install a current-generation web browser now. Learn more.
View related literature on analyzing sulfur compounds.
See Additional Product Details below for column configuration key.
Custom packed and micropacked columns also available by request.
Note: Initial 2" of column will be empty to accommodate a needle. For a completely filled column add suffix -901.
Agilent configuration also includes 1 1/2" void on detector side.
Sulfur analyses are traditionally performed using PTFE tubing to improve column inertness. Unfortunately, PTFE tubing is gas permeable, difficult to pack with high efficiency, prone to shrinkage, and has poor thermal stability. The Rt-XLSulfur packed or micropacked column eliminates these problems. The packing material for Rt-XLSulfur columns is extensively deactivated for analysis of low ppbv levels of hydrogen sulfide and methyl mercaptan. It is then treated to achieve effective separation of hydrocarbons from sulfur compounds. The interior wall and the end fittings of the Rt-XLSulfur column are Sulfinert treated, making the column as inert as PTFE. The extra care taken to manufacture this column ensures more accurate analyses of sulfur compounds.
Note: Columns do not include column nuts and ferrules. Optional installation kits can be ordered separately.
Column Configuration Key:
-800 General—General Configuration
-810 Agilent—(HP) 5880, 5890, 5987, 6890, 7980
-820 Scion (Bruker 430, 450) (Varian 3700, Vista Series, FID)
-830 PE/Sigma—PE 900-3920, Sigma 1, 2, 3
-840 PE Auto Sys—PE Auto System 8300/8400/8700, Clarus 500
-850 Shimadzu 14A—Shimadzu 14A, 2014
Other column configurations available.